|No.7 Sheng Feng Road, Xin He Industril Zone, Dong Guan City, Guang Dong Province, Chinafirstname.lastname@example.org|
|Place of Origin:||China|
|Minimum Order Quantity:||1set|
|Packaging Details:||Non-fumigation wooden box|
|Delivery Time:||25 working days|
|Payment Terms:||FOB Shenzhen|
|Supply Ability:||50 set per month|
|Pressure Range:||0.5～100Kpa, Commonly Used Test Pressure 11.6 Kpa||Inner Box Size:||500*600*500mm (width*height*deep)|
|Control Method:||PLC Man-machine Interface Control||Vacuum Display:||SMC Digital Display Vacuum Gauge|
|Timing Control:||Input Settings On The PLC Interface, Freely Set In 0-9999H (H/M/S Seco||Pressure Fluctuation:||≤5%|
|Pressure Preservation Time:||5% Reduction In 26 Hours||External Use Of Gas Source:||0.6~0.8MPA|
Product name: Battery Altitude Simulation Tester
Model number: GX-3020-ZC50
This equipment is used for battery high-altitude and low-pressure simulation test. The batteries are tested under the negative pressure of 11.6kPa (1.68psi) for 6 hours. The tested battery can not be exploded, no fire, no smoke, no leakage. To be qualified. The equipment can automatically control the test period (for example, 6H), and real-time monitor the pressure change in the tank, Can realize automatic termination of the test process.
UL 1642 "lithium battery standard"
UL 2054 "Household and Commercial Battery Pack"
UN38.3 "Recommendations on the Transport of Dangerous Goods - Manual of Tests and Criteria Part III"
MH / T 1052-2013 "Air Transport Lithium Battery Test Specification"
IEC62133-2012 "Safety requirements for batteries and accumulators containing alkaline or non-acidic electrolytes"
IEC 62281: 2004 "Safety Requirements for Lithium Batteries and Batteries in Transport"
IEC 60086: 2007 "Original battery Part 4 Safety requirements for lithium batteries"
1: The outer box is treated with high-grade paint of A3 steel plate. The coating is hard and firm, and has strong anti-rust ability;
2: The inner box is made of high quality SUS306# stainless steel plate material, and the fireproof, corrosion-resistant and easy-to-clean Teflon tape is placed in the box body to maximize the protection of the test box;
3: Filling material between inner and outer boxes: ultra-fine aluminum silicate cotton;
4: Double-layer tempered glass observation window is opened on the front of the cabinet, and the user can clearly observe the internal test situation of the box;
5: Automatically control the input and output of air, and automatically control the air pressure. The door seal is made of high temperature resistant silicone sealing strip, and the sealing effect is good to ensure the vacuum of the test box is constant;
6: Safety device: Built-in safety device with pre-overpressure, using pressure reducing valve to achieve over-pressure safety protection;
7: Alarm function: It will automatically alarm when the operation is wrong or after the timer ends.
8: Vacuum pump: Built-in vacuum pump, easy to operate and move;
9: Casters: Four universal wheels are mounted on the bottom for easy handling and movement.
|Inner box material||SUS304# First grade stainless steel, thickness 6mm|
|Outer box material||Cold rolled steel sheet baking treatment|
|Fill material||Ultrafine aluminum silicate|
|Pressure range||0.5～100Kpa, commonly used test pressure 11.6 Kpa|
|Inner box size||500*600*500mm (width*height*deep)|
|Overall size||800*1500*700mm (width*height*deep)|
|Storage partition material||SUS304# first grade stainless steel round bar|
|Observation window||Double tempered glass, size: 250*250*5 mm, internal glass with explosion-proof membrane, Square anti-projection net installed between double glass|
|control method||PLC man-machine interface control|
|Vacuum display||SMC digital display vacuum gauge|
|Timing control||Input settings on the PLC interface, freely set in 0-9999H (H/M/S seconds can be switched)|
|Pressure preservation time||5% reduction in 26 hours|
|Safety device||Overcurrent protection fuse, built-in pre-overvoltage safety device, use pressure reducing valve to achieve over-voltage safety protection, automatic operation or automatic alarm after timing|
|Equipment using power||3∮, AC220V±10%, 50Hz, 10A,|
|External use of gas source||0.6~0.8MPA|
Equipment use conditions
1 Avoid direct sunlight or high temperature and humidity;
2 The equipment is placed on a stable surface;
3. Please ensure that the ambient temperature is (20±5) °C
4. Air supply: 0.6 ~ 0.8 MPA
Main machine: 1 Set
Stainless steel shelf: 2 PCS
Manual: 1 Copy
Asbestos board: 1 PCS
Product certification: 1 Copy
Contact Person: Johnny Jiang