|No.7, Shengfeng Road 523039, Xinhe Industrial Park, Wanjiang, Dongguan, Guangdong, Chinaemail@example.com|
|Place of Origin:||CHINA|
|Minimum Order Quantity:||1 set|
|Packaging Details:||Non-fumigation wooden box|
|Delivery Time:||40 working days|
|Payment Terms:||T/T, L/C|
|Material:||Stainless Steel||Application:||Battery, Coin Cell, Pack|
|Control Model:||PLC Remote Control||Standards:||SJ/T11169, SJ/T11170, Ul1642, UL2054|
UL2054 PLC Remote Control Battery Pack External Fire Testing Machine
Battery pack external fire test machine for battery flammability test, according to SJ / T11169, SJ / T11170, Ul1642, UL2054 test standards. Battery pack external fire test machine battery tray run to the top of the fire, the fire rose to the main limit, the high fire timer arrived, the battery tray retracted to the limit battery tray retreat at the same time, the fire fell to the limit fire Disc cover before moving to the limit, ten minutes after the fire board cover retreat, in order to achieve the test standards.
1. The overall structure: the fire test device consists of the lower base, oil pan, cover moving rails, oil pan cover, battery placement bracket, console, battery pack lifting device
2. Oil pan structure: composed of stainless steel oil pan and oil pan bracket;
Oil pan material: made of stainless steel plate, the number and size of the oil pan set according to the demand side;
4. The overall framework black deal.
5. The control mode: plc remote control (equipment host and control cabinet connection cable length of 15m) + close-range adjustment
6. Guide rails: high temperature rail. oil pan fixed. mobile test mounting bracket; battery safety bracket and cover two guide rails. two per floor.
7. Load-bearing capacity: equipment base has sufficient strength. can take the weight of the battery pack 500kg and the weight of all structural parts of the upper part.
8. The bottom of the battery mounting bracket to the oil surface distance: the distance between the gasoline level and the test object adjustable in the range of 20cm ~ 70cm. with a height scale. common height of 50cm;
9. Battery mounting bracket can move distance: the furthest end and the oil pan distance of not less than 3.5m. the length of the moving rail is not less than 4m.
Battery mounting bracket moving speed: (6 ± 0.5) m / min
10. Oil spill collection tray: the bottom of the oil pan to take the steel floor to absorb oil spill. collection tray size can be included in the size of the oil pan. the number of two.
Contact Person: Annie Chen