ECE R100 Computer Control Fire Testing Equipment For Battery
|Place of Origin:
Payment & Shipping Terms:
|Minimum Order Quantity:
|30 Working days
|Fire Testing Equipment
ECE R100 Fire Testing Equipment,
Computer Control Fire Testing Equipment,
ECE R100 Battery Testing Equipment
Battery pack external fire testing equipment ECE R100 Battery Testing Equipment
This machine is mainly used for the safety verification test of PACK-level battery packs or systems and structural parts to evaluate the safety performance of the battery. The external fire of the battery pack simulates the occurrence of meta-explosion and ignition of the battery pack in a fire environment. Present the actual use status of battery packs or systems in the laboratory, discover and solve problems, optimize product design, and improve product quality.
System composition: The external fire test machine of the battery pack is composed of an oil pan, an oil delivery system, a cover plate, a battery mobile trolley, a test host platform, a control computer, a video monitoring system, a fire pool and a safety protection system (such as power protection, phase loss) Protection, overpressure/loss of pressure alarm protection).
Test standard: ECE R100
The main technical parameters:
|The test bench can bear the maximum weight
|The maximum size of the testable battery
|Dimensions of battery placement platform
|Dimensions of battery placement platform
|remote automatic ignition, automatic ignition is carried out on the remote console through an electronic ignition gun, and the fuel (gasoline) is at least 3m away from the test object during ignition
|computer operation control mode
|Control input mode
|program mode, fixed value mode
|water cooling system, test time, moving speed, etc. can be set; it can complete the fire test requirements in the latest international and domestic standards by inputting various variables.
|The overall structure of the fire station
|it is composed of the lower base, oil pan, cover plate moving guide, oil pan cover plate, battery placement bracket, console, battery pack hoisting device, fire extinguishing tank, etc.
|The burning table is fixed at the designated position, and the whole is divided into three layers (from bottom to top)
|The first layer is an oil pan, the size is not less than W3000*D2500mm, and it is fixed at the bottom of the equipment;
The second insulation layer is the fire extinguishing baffle, the size can cover the entire oil pan, and the fire extinguishing cover can be moved;
The third layer is the sample shelf, which can be moved as a whole.
|Sample shelf structure
|The whole adopts channel steel frame structure, which can load up to 1000kg, and has the characteristics of high temperature resistance, corrosion resistance, fire resistance, and calcination not easy to deform
|Battery pack placement support
|Through the upper lifting device, the height can be adjusted structure design, the distance between the battery pack or the system and the combustion plate can be adjusted according to the requirements, and the distance between the gasoline liquid level and the test sample can be adjusted.
|Oil pan movement mode
|The oil pan is fixed under the fire table, and the bottom is connected with the oil pipe. The oil pan does not need to be moved during the test. The advantage of this design is to avoid the risk of fuel liquid splashing out due to the oil pan being too large
|Movable method of fire extinguishing baffle
|control its movement by electric + manual + remote control, etc. The fire extinguishing baffle can be moved left and right on the fire bench, covering the oil pan for fire extinguishing
|The moving method of the sample shelving table
|the sample shelving table is used to install the test object, and can be moved to the top of the oil pan for fire test
|Specimen movement method
|during the test, it is realized by moving the test object (battery pack or system), that is, the test object is placed on the sample shelf, and the preheating, direct combustion, and indirect combustion are realized by moving the bench left and right. , End the test and other 4 steps
|The moving speed of the sample shelf
|1∮, AC220V, 50Hz, 20A